- Research article
- Open Access
- Open Peer Review
Use of real time polymerase chain reaction for detection of M. tuberculosis, M. avium and M. kansasii from clinical specimens
© Bainomugisa et al.; licensee BioMed Central. 2015
- Received: 27 December 2014
- Accepted: 1 April 2015
- Published: 14 April 2015
We’re sorry, something doesn't seem to be working properly.
Please try refreshing the page. If that doesn't work, please contact support so we can address the problem.
The incidence of M. tuberculosis (MTB) and non tuberculous Mycobacterium species (NTMs) like M. avium and M. kansasii has increased due to Human Immunodeficiency Virus (HIV) epidemic. Therefore accurate, rapid and cost effective methods for the identification of these NTMs and MTB are greatly needed for appropriate TB management. Thus in this study we evaluated the performance of Lightcycler® Mycobacterium detection assay to detect MTB, M. avium and M. kansasii in sputum specimens.
A total of 241 baseline minimally processed sputum specimens from individual adult TB suspected patients were analyzed by Mycobacterium detection assay (Real-time-PCR) on a LightCycler 480® while using liquid culture as a reference standard.
Real time PCR had a sensitivity of 100% (95% CI 96–100) and 100% (CI 19–100) for detection of MTB and M. avium respectively. Additionally the assay had a specificity of 99% (95% CI 96–99) and 95% (95% CI 91–97) for identification of MTB and M. avium respectively. The positive predictive value (PPV) for Real time PCR to identify MTB and M. avium among the specimens was 98% (95% CI 94–99) and 15% (95% CI 2–45) respectively. The kappa statistics for Real time PCR to identify MTB and M. avium was 0.9 (95% CI 0.9–1.0) and 0.3 (95% CI–0.03–0.5) respectively. The median time to detection for Real time PCR assay was 2 hours while overall median time to detection for MGIT-positive cultures was 8 days. The sample unit cost for Real time PCR was $ 12 compared to $ 20 for the reference liquid culture.
The Light cycler® Mycobacterium detection assay rapidly and correctly identified MTB and M avium thus has the potential to be adopted in a clinical setting.
- M. tuberculosis
- M. avium
- M. kansasii
- Real time PCR
Mycobacteria species cause a variety of illnesses including pulmonary tuberculosis (PTB) which has profound individual and public health implications . The absolute number of PTB cases occurring each year (9.4 million) is currently greater than at any time in history and the global incidence rate is estimated to have peaked . The continued rise of PTB may be largely attributed to the AIDS pandemic combined with weak healthcare delivery systems . Additionally, there is increase in the incidence of non-tuberculosis mycobacterial (NTM) disease in AIDS patients since the first cases in 1982 that complicates the disease . The commonest NTM’s associated with pulmonary infection among HIV patients are M. avium complex, M. kansasii, M. abscessus and M. fortuitum . The disease caused by M. kansasii often mimics pulmonary tuberculosis in signs and symptoms while M. avium causes disseminated disease. The identification of NTMs has lagged behind because of lack of infection awareness among physicians and microbiologists, lack of standardized criteria to define NTM respiratory disease and poor laboratory infrastructure to culture and identify NTMs. The lack of appropriate, rapid and accurate diagnostics tools remains critical and undermines progress towards the 2015 millennium development goals for TB control . Therefore, this calls for the need for evaluating more tools to use in accurate diagnosis of TB.
A number of tools have been used in TB diagnosis, for instance smear microscopy the most widely used test has low sensitivity especially in patients with extra pulmonary tuberculosis, those with HIV co-infection and TB due to NTMs . Nucleic acid amplification tests such as PCR based assay have great promise for TB diagnosis and rapid detection of drug resistance with commercial assays widely used in developed countries for over 20 years . However despite their simplicity, they are prone to PCR inhibitors, some tests require post amplification procedures that increases the turn-around time and some are limited by the DNA quantities in the starting material [9,10]. Furthermore, since most of these techniques require the isolates to be cultured first, this will introduce growth competition in cases of mixed infection and hence a selection bias. Recently WHO endorsed GeneXpert MTB/RIF for use in the diagnosis of TB in endemic countries  but it lacks markers for identification of M. avium and M. kansasii that are mainly associated with HIV patients. Therefore, the need for an alternative method that can comprehensively detect M. tuberculosis (MTB) and NTMs present in clinical specimens.
In this study we evaluated the LightCycler® Mycobacterium detection assay based on the principle of Real-time PCR technology for the detection of M. tuberculosis, M. avium and M. kansasii using species specific hybridization probes designed based on the 16S ribosomal RNA (rRNA) gene including the hyper variable region A [12,13]. We performed a clinical evaluation on this assay to estimate the cost effectiveness, turnaround time and analytical performance for TB diagnostic potential.
A total of 241 baseline sputum specimens collected as part of a standard patient care were randomly selected from clinical specimens sent to our TB laboratory between October 2012-February 2013.
The study protocol was reviewed and approved by the Institutional Review Board (IRB) at Joint Clinical Research Centre (JCRC). Individual informed consent was not sought because the study was conducted on routine samples only and it did not involve any intervention, additional samples or change in patient management. A patient consent waiver was approved by the IRB of JCRC.
Processing of sputum specimens
All specimens were decontaminated according to the available laboratory protocol . In brief, specimens were 1:1 mixed with N-acetyl-l-cysteine (NALC)-NaOH (final concentrations 1.5% NaOH, 0.7% NaCitrate, 0.25% N-acetyl-cysteine) and put on plat form shaker (Thermo scientific Inc. USA) at 60 rpm for 20 min. After neutralization with 0.5 M phosphate buffer (pH 6.8) and centrifugation (3000 × g for 20 min) in order to concentrate the mycobacteria, the sediment was re-suspended with 2 ml phosphate buffer.
Isolation of genomic DNA
In brief, 500 μl of the decontaminated sputum sample was spun and pellet re-suspended in 20 μl of nuclease free water; heat killed in a heat block set 95°C for 1 hour to lyse the bacteria and later sonicated at 37 kHz (Elma S30, Gottlieb-Daimler-Str. Singen, Germany) for 15 minutes. The resultant genomic DNA in the supernatant was recovered by centrifugation at 8000 g for 3 min for eventual use in the Real time PCR assay.
Reagents were aliquoted and each aliquot was used only once. Sterile microfuge tubes and 96 PCR well plates for Real time PCR assay use. Reagent preparation, DNA extraction, DNA amplification and detection were performed in separate rooms to avoid cross contamination of amplicons.
A positive control (Mycobacterium tuberculosis H37Rv, M. avium and M. kansasii) was included in each test and distilled water was included as a negative test control. Uracil-N-glycosylase (UNG) was used in the amplification process to avoid post PCR DNA contamination.
Mycobacterium real time PCR
Real time PCR was performed on a LightCycler® 480 II (Roche diagnostics, Mannheim, Germany) according to the manufacturer’s instructions using light cycler mycobacterium detection assay. This assay comprised of two major steps: (i) Amplification of internal control sequences and genomic target sequences (ii) melting curve detection by florescence measurement at 640 nm. A 20 μl reaction mixture contained 4 μl of sample lysate (or 4 μl of positive/negative control), 11 μl detection mix (primers and probes), 4 μl of PCR master mix (Taq polymerase, DNTPs, Mg2+), 0.75 μl internal control and 0.25 μl uracil-DNA gylcosylase. Thermal cycling was as follows: 10 min at 95°C, then 45 cycles of 10 sec 95°C, 10 sec 50°C and final extension of 20 sec at 72°C. Melting curve detection to determine the melting temperature (Tm) values for the target sequences was set as follows: 1 min at 95°C, 2 min at 40°C, 75°C continuous and then cooling at 10 sec at 40°C.
Analysis of data
The time that elapsed from MGIT sample inoculation and incubation to MGIT culture positivity or negativity was registered as the turnaround time (TAT) for liquid culture. The time difference between the start and stop time for each real time PCR run was used as the turnaround time for Real time PCR assay. The unit sample cost between Real time PCR and liquid culture use was achieved by comparing the requirements and their costs to test 241 clinical specimens for identification of M. tuberculosis, M. avium and M. kansasii.
Statistical data were entered and analyzed using Epi™ StatCal info version 7 software (CDC, Atlanta). The values got were validated using statistical diagnostic software MedCalc version 15.2.2 (MedCalc Software bvba, Belgium).
Analysis of the sputum specimens by Mycobacterium detection assay and liquid culture
Type of sample
Mycobacterium detection assay
RT PCR Negative
2x2 performance analysis of Real time PCR and Liquid culture
Mycobacterium detection assay
Performance of Mycobacterium detection assay
Performance of Real time PCR while using Liquid culture as the reference standard
Cost and time to detection
Costs comparison between Real time PCR assay and liquid culture
Mycobacterium detection assay
US cost ($)
US cost ($)
Bactec 960 MGIT culture media
Bactec 960 MGIT supplement
ZN carbolfuschin stain (BD)
ZN Decolorizer stain (BD)
Methylene Blue stain (BD)
Capilia TB Neo
Genotype mycobacterium CM kit
Light cycler Mycobacterium detection kit
US Unit cost ($) (241 samples)
6 hours; 4 minutes was the entire time to run 241 clinical specimens using Real time PCR assay in 6 runs giving an average time to detection for each run 1 hour; 40 minutes for identification of MTB, M. avium, M. kansasii or negative. The time to detection for negative liquid culture was 42 days (6 weeks) while that of positive cultures was 8 days (IQR 5.3–12.5).
Global TB control efforts are based on rapid diagnosis of disease cases followed by adequate treatment thus prevent continued transmission. With increased incidence of TB and non tuberculous disease infection especially among HIV patients, diagnostics with better sensitivity and ability to identify M. tuberculosis and non tuberculous mycobacteria are required for appropriate management. Among the commonest non tuberculous mycobacteria affecting HIV patients include M. avium and M. kansasii which can be genomic DNA amplified and specific products identified within the same reaction using florescence monitoring by Real time PCR. The recently WHO endorsed Real time PCR based assay; GeneXpert® (Cepheid, Sunnyvale, USA) that offers rapid identification of MTB and rifampicin resistance but cannot identify M. avium and M. kansasii. This study aimed at assessing Mycobacterium real time PCR assay to identify M. tuberculosis, M. kansasii and M. avium from patient clinical specimens at baseline diagnosis and the data shows that Real time PCR was sensitive in identifying MTB at baseline diagnosis and had a significantly short turnaround time compared to liquid culture.
By comparing Real time PCR with liquid culture as the reference standard, its accuracy in identifying mycobacterial species from sputum samples has been demonstrated in this study. Our study showed a high sensitivity (100%), specificity (99%) and positive predictive value (98%) for identification of MTB from baseline sputum specimens. Previous studies have described the use of Real time PCR in the analysis of sputum samples at baseline diagnosis and reported up 100% specificity for identification of MTB [15,16]. Therefore this Real time PCR assay is a suitable methodology for a clinician to take a decision when identifying MTB from baseline samples. This suggestion is further supported by the excellent agreement between Real time PCR and liquid culture for identification of MTB (kappa statistics, 0.9). However, Real time PCR assay may not be considered as a replacement for culture of MTB given the observation that it identified 1 specimen as MTB positive yet culture indicated it as negative. Clinical decision in the context of the patient may be important especially in the initiation of anti-tuberculosis therapy. Nevertheless, the possibility that missed identification of the specimen can be influenced by other factors such as DNA quality, DNA concentration, extraction method, salt on buffer solution among other factors that could affect amplification  cannot be ruled out. These were however not evaluated in this study.
Though this assay had high sensitivity and specificity for identification of M. avium from the specimens, it had a low positive predictive value (15%). This difference is brought about by the discrepancies in the ability of Real time PCR to identify certain species than HAIN genotype Mycobacterium CM; for instance Real time PCR identified 15 M. avium, of which only 2 were identified as M. avium by liquid culture while using HAIN genotype Mycobacterium CM reverse hybridization assay. This is further supported by the theory of M. avium cells being non-viable in liquid culture yet their DNA was identified by Real time PCR assay. Furthermore, one species of M. kansasii was identified by Real time PCR but could not be classified as a mycobacterium species by HAIN genotype Mycobacterium CM following liquid culture. Though another study with a larger sample size is needed to study this discrepancy, a study in South Africa reported M. scrofulaceum (human lymph node isolate) and M. flavescens to give a false positive signal for M. avium and M. kansasii respectively due to the similarity in the hyper variable region A of 16S RNA .
Mycobacterium real time PCR assay had an average time to detection (1 hour; 40 minutes) significantly lower than that of liquid culture which is comparable to other Real time technology like GeneXpert (2 hours)  This greatly reduces the time to initiation of anti-tuberculosis therapy and lost to follow-up cases due to delays in making diagnosis . In this study it was estimated that the unit specimen cost to test using reference standard was almost twice more expensive than use of Real time PCR assay, although the capital costs to buy the respective equipment was not included. In a South African nationwide feasibility study done by Boehme et al.  indicated that Real time PCR assays may be more expensive than smear microscopy but the costs were similar to sputum culture. A larger study may be needed to assess the cost effectiveness of this assay compared to liquid culture. Since this assay has amplification and detection done by the Roche 480 II instrument, there are limited chances of assay contamination than other assays like DNA line probe assays that involve further manipulations after amplification. With the demonstrated advantages of using this technique such as capacity to identify the M. tuberculosis and M. avium, high through put ability, cheaper cost than liquid culture and short turnaround time puts this Real time PCR assay at the forefront of molecular techniques to guide TB management especially in highly endemic mycobacterial diseased countries like Uganda.
The Mycobacterium Real time PCR assay correctly identified the majority of the culture confirmed M. tuberculosis with high specificity though identification of M. avium and M. kansasii needs to be assessed further especially in high risk population. The utility of this assay for TB diagnosis was comparable with liquid culture; thus it can be adopted in a clinical setting. This assay proved to be a rapid and cost-effective test compared to liquid culture for identification of M. tuberculosis and M. avium from clinical specimens.
This study was financially supported by the Forgarty International Centre (FIC)/National Insitutes of Health (NIH) grant number 2 U2R TW0006879 through the International Clinical Operational and Health Services Research (COHRE) at Joint Clinical Research Centre under; a collaborative training program implemented with Makerere University, Mbarara University of Science and Technology, Kampala City Council Authority, Case Western University and University of Georgia. Special thanks to the staff of Joint Clinical Research Centre-Mycobacteriology laboratory for their guidance and support.
- Tobin MJ. Tuberculosis, lung infections, and interstitial lung disease in AJRCCM 2000. Am J Respir Crit Care Med. 2001;164(10 Pt 1):1774–88.View ArticlePubMedGoogle Scholar
- WHO: Global Tuberculosis Report. In. Edited by Data WLC-i-P: World Health Organisation; 2014: 1–134.Google Scholar
- Corbett EL, Marston B, Churchyard GJ, De Cock KM. Tuberculosis in sub-Saharan Africa: opportunities, challenges, and change in the era of antiretroviral treatment. Lancet. 2006;367(9514):926–37.View ArticlePubMedGoogle Scholar
- Gopinath K and Singh S. Non-tuberculous mycobacteria in TB-endemic countries: are we neglecting the danger? PLos Neglected Tropical Diseases. 2010; 4 (4).Google Scholar
- Johnson MM, Waller EA, Leventhal JP. Nontuberculous mycobacterial pulmonary disease. Curr Opin Pulm Med. 2008;14(3):203–10.View ArticlePubMedGoogle Scholar
- Lawn SD. Pre-screening with GeneXpert (R) MTB/RIF may increase use of isoniazid preventive therapy in antiretroviral programmes. Int J Tuberc Lung Dis. 2011;15(9):1272–3. author reply 1273–1274.View ArticlePubMedGoogle Scholar
- World Health Organization. Laboratory services in TB control, part I: organization and management, In., vol. 1. Geneva: WHO; 1998.Google Scholar
- Pai M, Minion J, Sohn H, Zwerling A, Perkins MD. Novel and improved technologies for tuberculosis diagnosis: progress and challenges. Clin Chest Med. 2009;30(4):701–16. viii.View ArticlePubMedGoogle Scholar
- Albert H, Bwanga F, Mukkada S, Nyesiga B, Ademun JP, Lukyamuzi G, et al. Rapid screening of MDR-TB using molecular line probe assay is feasible in Uganda. BMC Infect Dis. 2010;10:41.View ArticlePubMedPubMed CentralGoogle Scholar
- Teo J, Jureen R, Chiang D, Chan D, Lin R. Comparison of two nucleic acid amplification assays, the Xpert MTB/RIF assay and the amplified Mycobacterium tuberculosis direct assay, for detection of Mycobacterium tuberculosis in respiratory and nonrespiratory specimens. J Clin Microbiol. 2011;49(10):3659–62.View ArticlePubMedPubMed CentralGoogle Scholar
- World Health Organization. Automated real-time nucleic acid amplification technology for rapid and simultaneous detection of tuberculosis and rifampicin resistance: Xpert MTB/RIF system. Geneva, Switzerland: World Health Organization; 2011.Google Scholar
- Light cycler mycobacterium detection kit, version April 2009; Roche Diagnostics Gmbh, Mannheim, Germany.Google Scholar
- Omar SV, Roth A, Ismail NA, Erasmus L, Ehlers M, Kock M, et al. Analytical performance of the Roche LightCycler (R) Mycobacterium detection kit for the diagnosis of clinically important mycobacterial species. PLoS One. 2011;6(9):e24789.View ArticlePubMedPubMed CentralGoogle Scholar
- Kubica GP. Phage typing of Mycobacterium tuberculosis: a time for standardization. Am Rev Respir Dis. 1982;126(1):3–4.PubMedGoogle Scholar
- Richardson ET, Samson D, Banaei N. Rapid Identification of Mycobacterium tuberculosis and nontuberculous mycobacteria by multiplex, real-time PCR. J Clin Microbiol. 2009;47(5):1497–502.View ArticlePubMedPubMed CentralGoogle Scholar
- Nasr Esfahani B, Rezaei Yazdi H, Moghim S, Ghasemian Safaei H, Zarkesh Esfahani H. Rapid and accurate identification of Mycobacterium tuberculosis complex and common non-tuberculous mycobacteria by multiplex real-time PCR targeting different housekeeping genes. Curr Microbiol. 2012;65(5):493–9.View ArticlePubMedGoogle Scholar
- Perng CL, Chen HY, Chiueh TS, Wang WY, Huang CT, Sun JR. Identification of non-tuberculous mycobacteria by real-time PCR coupled with a high-resolution melting system. J Med Microbiol. 2012;61(Pt 7):944–51.View ArticlePubMedGoogle Scholar
- Sekadde MP, Wobudeya E, Joloba ML, Ssengooba W, Kisembo H, Bakeera-Kitaka S, et al. Evaluation of the Xpert MTB/RIF test for the diagnosis of childhood pulmonary tuberculosis in Uganda: a cross-sectional diagnostic study. BMC Infect Dis. 2013;13:133.View ArticlePubMedPubMed CentralGoogle Scholar
- Ortu S, Molicotti P, Sechi LA, Pirina P, Saba F, Vertuccio C, et al. Rapid detection and identification of Mycobacterium tuberculosis by real time PCR and bactec 960 MIGT. New Microbiol. 2006;29(1):75–80.PubMedGoogle Scholar
- Boehme CC, Nicol MP, Nabeta P, Michael JS, Gotuzzo E, Tahirli R, et al. Feasibility, diagnostic accuracy, and effectiveness of decentralised use of the Xpert MTB/RIF test for diagnosis of tuberculosis and multidrug resistance: a multicentre implementation study. Lancet. 2011;377(9776):1495–505.View ArticlePubMedPubMed CentralGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.